AUTOOL LM120+ Digital Refrigeration Manifold can be set according to the type of refrigerant that the car needs to fill; The refrigerant filing status can be observed through a visible window. The intelligent digital display of the test results of the pressure and temperature of the air conditioning system reduces the cost of time.
R11 R13 R113 R114 R115 R116 R12 R123 R124 R125 R1270 R134A R14 R141B R142B R143A R152A R170 R21 R218 R22 R227EA R23 R236EA R245CA R245FA R290 R32 R401A R401B R401C R402A R402B R403A R403B R404A R404B R405A R406A R407A R407B R407C R407D R407E R408A R409A R41 R410A R410B R411A R411B R412A R413A R414A R414B R415A R415B R416A R417A R418A R419A R420A R421A R421B R422A R422B R422C R422D R423A R424A R425A R426A R427A R428A R50 R500 R501 R502 R503 R504 R507A R508A R508B R509A R600 R600A R717 R744(Co2) R1234. Pressure test unit: Kpa; Mpa; bar; inHg; PSI. Pressure test range: 0 Kpa - 6000 Kpa.
Pressure test resolution: 1 Kpa. Pressure test accuracy: +/- 0.5 %(FS)+ 5dgt. Pressure overload limit: 10000 Kpa (10 Mpa; 100 bar;).
Vacuum test unit: Kpa; Mpa; bar; inHg; PSI. Vacuum test range: -101 Kpa - 0 Kpa.
Vacuum test resolution: 1 Kpa. Temperature test unit: °C (Celsius), °F (Fahrenheit). Temperature test range: -40°C-150°C (-40°F-302°F). 0.1°C (-40°C-99.9°C), 1°C (100°C-150°C). 0.1°F (-40°F-99.9°F), 1°F (100°F-302°F).Temperature test accuracy: +/- 0.5 °C + 2dgt +/- 0.9 °F + 2dgt. Built-in 89 kinds of refrigerant NIST: According to American NIST standard. Power Supply: 4 X 1.5V SIZE. AA / LR6 Battery not include. 1x AUTOOL lm120+ car air conditioning refrigerant main unit. 1x User Manual (in English). LM120+ safety standards for design and production. 1 The pressure measured by the digtal manifold pressure tester is gauge pressure. 2 Pressure testing ranges from -101Kpa to 6Mpa (-0.1bar to 60bar). 3 The limit pressure is 10 Mpa (100 bar).
4 The maximum operating pressure of standard hose is 600 PSI (approximate 4.13 Mpa, 41.3 bar). The limit pressure is 3000 PSI approximate 20.68 Mpa, 206.8 bar. 5 Please confirm the rated pressure value of the tested equipment before testing. Do not use it if it exceeds the range of the instrument.Also for DY517A, if the packed hoses does not match the pressure requirement, you can use suitable replacements for testing. 6 Do not use and store the instrument in high temperature, high humidity, flammable, explosive and strong electromagnetic fields. 7 Please do not change the instrument internal circuit, to avoid any damage of the instrument or danger occurring. 8 Please wear qualified protective equipment to protect user during testing. 9 Please use the instrument in a well-ventilated environment to prevent inhalation of toxic gases. This item is in the category "Business & Industrial\HVAC & Refrigeration\HVAC & Refrigeration Tools\HVAC & Refrigeration: Gauges & Probes". The seller is "autool_official_store" and is located in this country: US.
This item can be shipped to United States.